Oxygen delivery failure due to improper installation of carbon dioxide absorbent canister: a case report
The CLIC system in the Dräger Apollo anesthesia workstation allows a successful pre-use machine checkout without the presence of a carbon dioxide absorbent canister. It also allows the canister to be changed without interrupting controlled ventilation. However, this canister can be easily installed improperly with the CLIC adapter. We report a case in which a patient could not be ventilated by mask after the induction of general anesthesia, resulting in oxygen desaturation before successful ventilation was achieved with a bag valve mask. This case illustrates the importance of a leak test after components of the breathing circuit are changed.
1 Drager Just “CLIC” it! (2015) Available at: https://www.draeger.com/Products/Content/lsa_draegersorb_clic_casestudy_9051467_en.pdf. Accessed 2019 Apr 22
2 Drager Advanced Anesthesia Solutions: Drager Apollo Anesthesia Workstation (2017) Available at: https://www.draeger.com/products/content/apollo-br-9050390-us.pdf. Accessed 2019 Apr 22
3 SA Ianchulev, ME Comunale To do or not to do a preinduction check-up of the anesthesia machine Anesth Analg, 101 (2005), pp. 774-776
4 American Society of Anesthesiologists ASA Recommendations for Pre Anesthesia Checkout Available at: https://www.asahq.org/standards-and-guidelines/2008-asa-recommendations-for-pre-anesthesia-checkout. Published 2019. Accessed Apr 22, 2019 (2008)